Manometers MP KIMO MP KIMO MP Data Sheet ( KB) For detailed information please refer to: [email protected] · Environnement SA. This is the MP Thermo-Anemo-Manometer. It is an easy to use meter which can take complex measurements. The MP is designed to measure Air. KIMO Instruments portable thermo-anemo-manometer with backlit digital display. Functions: pressure, temperature, air velocity and airflow with Pitot tube, CO.
|Published (Last):||22 November 2011|
|PDF File Size:||16.25 Mb|
|ePub File Size:||14.33 Mb|
|Price:||Free* [*Free Regsitration Required]|
Kimo MP 200 P User Manual
Probe display Available information are: Create or launch a continuous dataset Select dataset A continuous dataset can be carried out using MP and is composed of several dated measuring points.
The numbers of points is displayed. This datasets can’t be set using Datalogger Software.
Select compensation It is possible to modify the value of the compensation in temperature. The ”Probe” klmo only appears when probes or module are connected. Select Start for dataset launching. Press Ok for launching measurement. By pressing ‘print’ button you will print only current page.
Kimo MP Multi Function Manometers | Pullman Instruments
It is thus necessary to enter the real temperature of use to obtain more precise results. In regards of Quality insurance norms, we recommend that the instruments are checked once a year.
This menu allows to view probe information plugged to C2, Module, C1 or wireless connections. When CO peak is selected, the period is diplayed 30 seconds by default. Create or launch a continuous dataset Select dataset A continuous dataset can be carried out using MP and is composed kkimo several dated measuring points.
You can view each measuring points if you click on Visu.
Click on Mesure to display values table of selected dataset. Launch a planned dataset A planned dataset is composed of several locations.
KIMO MP manometer buy in Kamenskoe
You can choose between a planned or a kimmo dataset. With Pressure function, you can access to following sub-functions – Hold – see Air velocity – Config.
To print the manual completely, please, download it. The operator can choose an automatic or a manual dataset, with an instant value or an average.
Comments to this Manuals Your Name. If K factor is selected, you must enter value. When the solenoid valve is enabled, it runs every minute. Configuration – Params Parameters – Avg. If you click on average icon, max. Planification must be made via the software. You can choose between a planned or a continuous dataset. From this screen you can: You must enter duration: Alarm threshold corresponds to the right value on the bargraph.
When the countdown is finished, the CO max is displayed. Don’t show me this message again. When the measurement is fixed, press Min Max to access to maximum and minimum values of the measurement.
Several actions are possible during the measurement: